Blood glucose test:
The Clover A1c can test the blood glucose level by simply connecting the Glucose Module.
Transfer test results to a computer:
Use the PC connection Cable (USB) to download and manage the test results via a computer.
Printing test results with the thermal printer:
When the Clover A1c Analyzer LCD display is showing the test result, simply press the printer button and the test result is printed by the thermal printer.
The barcode scanner can be used to scan the patient's information. All the patient's information is stored in the analyzer for future reference.
Infopia Clover A1c - Principle of a test: The Clover A1c test cartridge includes a cartridge and a reagent pack containing the reagents necessary for the determination of haemoglobin A1c, with a collection leg for blood sample collection. The reagent pack is pre-filled with a reaction solution and a washing solution. The reaction solution contains agents that lyse erythrocytes and bind Hemoglobin specifically, as well as a boronate resin that binds cis-diols of glycated Hemoglobin. A blood sample of 4µl is obtained at the collection leg of the reagent pack. The reagent pack is inserted into the cartridge where the blood is instantly lysed releasing the Hemoglobin and the boronate resin binding the glycated Hemoglobin. The cartridge is inserted into the Clover A1c Analyzer. The blood sample mixture is rotated to the measurement zone of the cartridge, where the amount of total haemoglobin is the blood sample is measured by the reflectance of the photo sensor LED (Light Emitting Diode) and PD (Photo Diode). Then the assembled cartridge is rotated so that the washing solution washes out non-glycated Hemoglobin from the blood sample, thus the amount of glycated Hemoglobin can be photometrically measured. The ratio of glycated Hemoglobin with respect to total Hemoglobin in the blood sample is calculated.
|Shop by brand||No|